- Model NO.: SDT-XR11
- Certification: ISO, CE
- Material: Metal
- Medical Devices Reg./Record No.: Lsyjxzz-2013-2300442
- Tube Voltage: 70kv
- Trademark: Hainuo
- Specification: 1set/carton
- Applicable Departments: Oral Rehabilitation Department
- Type: X-Ray Equipment
- Medical Device Regulatory Type: Type 2
- Power: 480W
- Tube Current: 8mA
- Transport Package: Carton / Wooden Case
- Origin: China
Using the international advanced technology,efficient integrated design,high efficiency,low radiation.
Using microcomputer intelligent control,not only remote controlled exposure,but more powerful function of low voltage alarm and high voltage protection.
Micro focus technology,much more clear image and accurate diagnosis.
The base has two modes of mobile and fixed ,pneumatic lift soat ensures more convenient for dentist and more comfortable for patient.
Can use lightroom dental film,imaging in one minute,maximum convenient dentist to diagnose.
Can connect to dental digital imaging system,that is indispensable for clinicians to diagnose and root canal filling.
Single phase AC 220V±22V; 50/60Hz
Power capacity: >1.1kVA
Power resistance: <2Ω
Tube voltage: 70kV
Tube current: 8mA
Total filtration: 2.5mmAL
Loading time: 0.2~4.0s, following R'10 series
Focus spot to image receptor distance: 210mm
Horizontal rotating angle of arm: >40°
Up-down range of arm: >400mm
Leak radiation: outside one meter<=0.002mGy/h
Unit weight: 50kg
Package size: 152*57*26(cm)
Net weight: 57kg
1. In order to ensure safety in operation and use, good ground lead must be installed well.
2. Because small X-ray machine's thermal capacity is limited, attention should be paid to making the machine work intermittently.
3. Only during exposure process this machine will produce X-rays. It won't produce X-rays in power-off state, neither in power-on state as long as not in exposure.
4. If any unforeseen circumstances occur during correct use, operators should cut off the power immediately and stop using the machine. It cannot be used again until it is overhauled and qualified.
5. Before leaving the factory, each X-ray device is equipped with an additional filter plate with the following specifications: 1.0/0.5mm aluminum sheets.
6. Every time when disassembling the machine, disassemble the head first and then other parts.
8. Every time after using the machine, recover the cross arm to the highest point, and then place the machine head in an equilibrium position.
9. Every time after using, sterilize and clean beam tube end that contacts the patient with medical alcohol.
10. Clean the machine surface with a dry cloth every week and inspect whether the connectors contact well.
11. Pay attention to machine maintenance. Check whether the indicator lights, display, rotary knob, switch, and remote control work normally; whether the X-ray source modules leak oil; whether when X-ray source modules expose produces any abnormal noise; or whether fixed screws in the mechanical part loose, and the like.
13. The rated capacity of the machine is 1.1kw. Please select power supply or power grid with a power supply capacity greater than 1.5kw. If network voltage in this region is abnormal, we recommend matching 1.5KW or higher industrial voltage stabilizer, to prevent damage to the machine and reduced machine life due to abnormal voltage.
Why Smile Dental?
8 years special in Dentistry
2. Supply Range
From equipment to materials
Form disposables to consumables
For dental clinic and dental laboratory
We offer almost all kinds of dental products.
From world's top brands
And wise choices of Chinese manufacturers based on 8 years' experience
Full certifications as ISO, CE, FDA
We never forget our positioning.
Long-term & stable warranty
With 100+ countries' transportation experience
With 3000+ KG monthly shipment amount via express courier
With 360+ CBM yearly shipment amount by sea
With professional & skilled transport operations
We deliver a smooth shipping and logistics experience at competitive cost.
Cost-effective product + service
Discount for bulky order
We stock leading brands as following: